|Performance:||>30 lpm clinical airflow, <50 mmHg to >525 mmHg vacuum|
|Dimensions:||9" H x 5.25 W x 17 L (22.9 cm H x 13.3cm W x 43.2cmL)|
|Weight:||10.05 lbs/4.55 kgs|
|Charger:||Internal dual mode charger activates when connected to AC power, dropping back to float when battery reaches a full charge. LED's indicate charging status and remaining running time.|
|Battery:||12v DC Sealed Lead Acid Battery|
|Run Time:||45 minutes +/- 10% when battery fully charged at full capacity|
|Input:||100-240V AC, 1.0-0.5A, 50-60 Hz|
|Output:||40 watts 12V DC 3.3A|
Canister, 2m patient tube, HI-D suction tip, AC power cord and Quickstart Guide.
ETL listed and conforms to ISO 10079-1, UL 60601-1, IEC 60601-1, and EN 60601-1-2 and certified to CAN/CSA STD C22.2 NO. 601.1-M.90, IP34 per IEC 60529, EN1789:1999.
This device contains a Sealed Lead Acid Battery. Sealed Lead Acid Batteries contain lead and lead compounds, chemicals known to the State of California to cause cancer and reproductive harm. Batteries also contain other chemicals known to the State of California to cause cancer. For more information, go to www.P65Warnings.ca.gov.